Juan Pablo Posted August 9, 2022 Report Share Posted August 9, 2022 The magnet is held by turning and has an edge of just a few mm of plastic. This edge has broken after 1 year and 2 months. My theory is that it has broken due to the same force that the magnet exerts when removing the implant from our heads, it exerts a force contrary to ours and the plastic edge is what resists those forces, it finally broke. I think it's a manufacturing defect, so I took it to MedEl today and they told me they're going to check it out. Has anyone else had a similar problem?magnet socket has broken Quote Link to comment Share on other sites More sharing options...
Join the conversation
You can post now and register later. If you have an account, sign in now to post with your account.